Page 62 Problem Description: Table Of Contents Status Icons.. If the camera is idle for 1. Insert the camera or battery into the charge station. For information on how to use a third party imaging software for image acquisition, see the Acquiring Images Through Third Party Imaging Software section.
|Date Added:||2 December 2007|
|File Size:||31.1 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Used Dental Intraoral Cameras | Kodak 1500
Page 43 USB port located at the back of the computer. Page 38 Figure 19 Active Camera Note: Capture Window The Capture window icon is located in the upper-left corner of the screen. Page 49 The Acquisition panel switches to the Review window, and the captured image is displayed in the Preview screen.
Setting up the Docking Station Connecting the Docking Station to a Computer To connect the docking station to a computer, follow these intraodal Remove and dispose of the protective sheath when image acquisition is completed. Power indicator Intraorwl indicates the camera is powered on; no light indicates the camera is powered OFF. TWAIN limitations are the following: Live video does not work.
Make sure the cable is locked into place. Settings Displays user-adjustable settings for the acquisition module and camera. Comment button Displays the comment and tooth number functions.
Acquiring Images Using A Computer When image acquisition is completed, remove and dispose of the protective sheath. Wait until the connection status turns to ‘Connected’.
Such action may cause an electric shock or a malfunction of the unit. Computer System Requirements Direct X 9. Page 66Inc. For information on how to use a third party imaging software for image acquisition, see the Acquiring Images Through Third Party Imaging Software section.
Battery Insert rechargeable cylindrical Lithium-ion battery into bottom of camera until it fits tightly in place. Carestream Cmaera is a trademark of Carestream Health, Inc.
Charge Station Components The charge station is used to charge the wireless camera and the additional battery simultaneously or separately. When image acquisition is completed, remove and dispose of the protective sheath. intraorzl
Carestream Kodak Dental Intraoral Camera Twain DEXcam Gendex Sopro | eBay
The camera reverts to the sleeping mode when it is the charge station. The following figures illustrate the idle and camerq camera status. Connecting the Charge Station to the Docking Station To connect the charge station to the docking station, follow these steps: Status Icons When using the camera and the docking station with a video monitor, the following icons are displayed on the video monitor to indicate equipment status. Vamera to the computer connected to the active docking station where work will be performed.
Select the “Activate Camera” button within the acquisition module. Table of Contents Add to my manuals Add. Be sure to note the color of the indicator if the camera does not work correctly.
Disposing Of The Battery Your community may offer a battery collection program. Page 40 – Connecting the Charge Station to the Doc Click the Settings tab.
The power indicator on the front of the docking station turns green.
CS from Carestream Dental | Dentalcompare: Top Products. Best Practices.
Comments to this Manuals Your Name. To record a video, click the button in the Capture window.
Page 10 WiFi indicator Solid blue indicates the connection is working properly; blinking blue indicates a communications problem.